Cautions During Treatment
1.Don’t operate the machine on the electriferous floor.
6.The electrode probe is moved on the skin by the circulation way, but don’t stop on one part .
4.The operator and customer DO NOT be with the metal decoration during in treatment time.
5.The plane of electrode probe must fully cling to the skin in the treatment time, but it can’t be
slant in use, because the electrode probe head doesn’t correctly touch the skin, which will
produce uncomfortable feeling and burn the skin.
2.Other people DO NOT touch the electrode flat when the machine is operating.
3.Other people DO NOT touch the operator and customer during treatment time.
6.People with metal inside the body(include metallic tooth)
5.Medical Plastic parts, or parts with artificial in filling inside.
7.Patient with skin inflammation or with edema.
8.People with an abnormal immune system.
9.Numb or insensitive to heat.
4.Pregnant women and child.
3.People with hemorrhagic disease, trauma or who is bleeding.
2.Patients with acute inflammation, asthma, deep vein thrombosis, thyroncus, cancer.
1.People with heart disease or high blood pressure, or configured cardiac pacemaker.
Input Voltage: AC220~230V 50/60Hz
Output Power: 1500VA
Fuse: FL5*2010A with buffer
Contact us if you are interested.